Yes you can. See also: Test Type Working Principles
Audiqueen comes with a powerful and comprehensive speech audiometric module that allows to perform speech audiometry with words or sentences, in quiet or in noise, with manual or adaptive scoring.
You can use existing speech lists or create your own.
The basic idea is that you connect the audio output of your PC soundcard (either built-in or external) to the AUX input of your audiometer. The latter must be correctly calibrated by the audiometer-technician. If this is well done, you only need to calibrate the connection between your PC and the AUX input. See Solution Centre for instructions for supported or connected/generic audiometers.
Select Otoconsult’s built-in '(Speech) Audiometric Module' in the format tool. Then click on New Measurement in the ribbon.
Select the audiometer or one of Audiqueen’s audio devices to deliver the sound.
Make sure that the connection between Audiqueen and this audiometer is correctly calibrated !!
The Speech Audiometry windows display all typical functions needed for speech audiometric testing.
Speech Audiometry Interface.
Set the test parameters (1), such as the speech list, the side to be tested, the conduction (free field, headphones, etc.), the presentation level of the signal and the noise (if relevant), etc. Don't forget to use the right mouse button (on the headphone or speaker icon) to set more parameters, such as masking, monitor and talk back, etc...).
Click ‘start’ (2) to play the first list.
This list order is shown in (3a) and the item order of the active list is shown in (3b). Either list or item order can be changed by (un)checking the corresponding shuffle checkbox.
|Shuffle Lists randomly changes the order of the different lists within one speech audiometry.|
Shuffle Items randomly changes the order of the different items within one speech list.
The permutate checkbox creates completely new lists with randomly chosen items across all existing lists (see this white paper on its clinical validation).
Scoring (4) is done per segment or item according to the settings of the test.
Have a look at the following video’s for an example of a speech audiometry test:
|Typical Procedure||Advanced Features|
A number of settings can be modified in the configuration window .
Here you can choose how to score the segments, whether to highlight one or more intensities, whether to convert the SPL axis to HL and whether to play video segments.
Here you can set the intensity resolution and the initial levels for the signal and the noise.
Here you can manage the adaptive methods. See Adaptive method for details.
See this separate support page for alle details.
If the soundcard is set to be used with more than 2 speakers, you will be able to configure through which speakers sound should be played to present the stimulus signals and the noise.
|This feature is not available in the webviewer. |
Audiqueen's webviewer displays the results entered in Audiqueen's Workstations.